Cold and deep light
Technical datas and Employ instructions. User manual.
Manufacturer : RIMSA P.LONGONI srl - Seregno (Mi) Italia
model : A/50-M
Power connection : 230V ~ (current)
Nominal frequency : 50/60 Hz
consumption : 55 W
primary fuse : F 1A.
secondary fuse : T 2A.
classification: Class I
level protection against direct and indirect contact : type B
level protection against liquid entrance: IP20
bulb : halogen 50 W - 24 V. GY6,35 with aluminium socket
guarantee : 12 months from delivery date
Medical device classification: class I - 93/42 CEE
Rimsa Lamps are sold assembled and packed in cartoons box.
After taking the lamp out of the packing, proceed as follows:
A) mod. A/50-M with caster base:
1-insert the pivot of the lamp into the hole upper the caster base.
2- screw the black screw to avoid the lamp structure comes out the tube
3-insert the eletric plug, which is attached at the end of electric cable, in the
4-turn-on the lamp by pushing the switch .
B) mod. A/50-M wall type (wall clamp S/12)
1-fixing the clamp S/12 to the wall with no. 3 expansion screws.
2-insert the lamp into the hole of the wall clamp S/12.
3-see points 2, 3 and 4 above mentioned
Before any maintenance, such as substitution of bulbs or fuses, log out the lamp and remove the cable
from the ground connection.
1-remove the 2 screws positioned on sides of the aluminium reflector
2-remove the plexiglass protection
3-substitute halogen bulb 50 W 24 V. GY6,35 with aluminium socket (use original spare parts Rimsa)
4-put the plexiglass to original position and screw the two screws on sides of reflector.
Clean and maintenance
The apparatus does not require periodic maintenance, only replacement of bulb when fails .
To clean the lamp use a damped cloth with disinfectant.
The manufacturer is responsible for the safety of the structure, only if:
the assembling, addings, changes or fixings are performed by authorized personnel and if the power
installation of the room respects the IEC prescriptions that could be enforced: and at last, if the lamp is used
according to the employ instructions.
=level protection against direct and indirect contact: Type B
=consumption of the lamp
=All electrical and electronic products should be disposed of separately from the
municipal waste stream via designated collection facilities appointed by the government or the
DICHIARAZIONE DI CONFORMITA' DEL COSTRUTTORE
DECLARATION OF CONFORMITY OF THE COMPANY
RIMSA P. LONGONI S.R.L.
The company: Via Monte Rosa, 18-20-22 - 20038 SEREGNO (MI) - ITALIA
dichiara che il seguente apparecchio di illuminazione:
declare that the following model:
50W-24V GY6,35 cl.I, IP20
a cui questa dichiarazione si riferisce, è conforme alle seguenti norme di
to which this declaration relates conforms with the following standard:
♦ EN 60601 - 1 (II Ed.) + A1+A2+A11+A13 (Norma generale,)
♦EN 55015 (1993)
ed ai requisiti delle seguenti direttive CEE:
in accordance with the provisions of:
DIRETTIVA DISPOSITIVI MEDICI 93/42 CEE
MEDICAL DEVICE DIRECTIVE 93/42 CEE
DIRETTIVA EMC 89/336 CEE
EMC DIRECTIVE 89/336 CEE
Luogo e data
(Place and date)
Timbro e firma del responsabile
(Signature or valid mark of the authorized person)
CERTIFICATE OF GUARANTEE
1. RIMSA structures are guaranteed for 12 months as of delivery day.
2. During the period of guarantee, RIMSA will fix and/or substitute all defective
pieces, when it's well verified that they are due to manufacturing causes.
3. The Guarantee does not include the components subject to wear, such as
halogen bulbs and fuses.
The Guarantee does not include:
•breakdown or defects due to installation errors.
•breakdowns or defects due to negligence and abnormal use of the product.
•technical interventions for presumed defects or unnecessary tests.
The user loses the right to the Guarantee in case that evidence is found of
interventions or tampering from personnel not authorized by RIMSA assistance